Research comparing portable body composition methods, such as bioelectrical impedance analysis (BIA), to air displacement plethysmography (ADP) is limited. We assessed reliability and validity of predicting fat-free mass (FFM) by the RJL, Omron, and Tanita BIA machines using ADP via BodPod as a criterion. FFM (kg) was assessed twice in college students (N = 77, 31 males and 46 females; age = 19.1 ± 1.2 years) using ADP, RJL, Omron, and Tanita BIAs. Reliability was assessed using analysis of variance to obtain an intraclass correlation statistic (Rxx). Validity was assessed using Pearson correlation (r) coefficient. FFM averaged 75.6 ± 9.4 kg in men and 59.8 ± 7.6 kg in women. Reliability was high in both genders RJL (Rxx = .974–.994), Omron (Rxx = .933–.993), and Tanita (Rxx = .921–.991). Validity within males was also high: RJL (r = .935), Omron (r = .942), and Tanita (r = .934), and only slightly lower in females: RJL (r = .924), Omron (r = .897), and Tanita (r = .898). The RJL, Omron, and Tanita BIA machines appear to be both reliable and valid for predicting FFM of male and female college students. Therefore, any of these three BIA devices is appropriate to use for body composition assessment in a healthy adult population.
Vasold, Parks, Phelan, Pontifex, and Pivarnik are with the Dept. of Kinesiology, Michigan State University, East Lansing, MI.
FieldsD.A.HunterG.R. & GoranM.I. (2000). Validation of the BOD POD with hydrostatic weighing: Influence of body clothing. International Journal of Obesity and Related Metabolic Disorders24(2) 200–205. PubMed ID: 10702771 doi:10.1038/sj.ijo.0801113
Fields, D.A., Hunter, G.R., & Goran, M.I. (2000). Validation of the BOD POD with hydrostatic weighing: Influence of body clothing. 24(2), 200–205. PubMed ID: 10702771 doi:10.1038/sj.ijo.080111310.1038/sj.ijo.0801113)| false
FornettiW.C.PivarnikJ.M.FoleyJ.M. & FiechtnerJ.J. (1999). Reliability and validity of body composition measures in female athletes. Journal of Applied Physiology87(3) 1114–1122. PubMed ID: 10484585 doi:10.1152/jappl.1918.104.22.1684
KhaledM.A.McCutcheonM.J.ReddyS.PearmanP.L.HunterG.R. & WeinsierR.L. (1988). Electrical impedance in assessing human body composition: The BIA method. The American Journal of Clinical Nutrition47(5) 789–792. PubMed ID: 3364394 doi:10.1093/ajcn/47.5.789
LadduD.R.LeeV.R.BlewR.M.SatoT.LohmanT.G. & GoingS.B. (2012). Predicting visceral adipose tissue by MRI using DXA and anthropometry in adolescents and young adults. International Journal of Body Composition Research10(4) 93–100. PubMed ID: 26097436
Laddu, D.R., Lee, V.R., Blew, R.M., Sato, T., Lohman, T.G., & Going, S.B. (2012). Predicting visceral adipose tissue by MRI using DXA and anthropometry in adolescents and young adults. 10(4), 93–100. PubMed ID: 26097436)| false
LukaskiH.C.JohnsonP.E.BolonchukW.W. & LykkenG.I. (1985). Assessment of fat-free mass using bioelectrical impedance measurements of the human body. The American Journal of Clinical Nutrition41(4) 810–817. PubMed ID: 3984933 doi:10.1093/ajcn/41.4.810
Lukaski, H.C., Johnson, P.E., Bolonchuk, W.W., & Lykken, G.I. (1985). Assessment of fat-free mass using bioelectrical impedance measurements of the human body. 41(4), 810–817. PubMed ID: 3984933 doi:10.1093/ajcn/41.4.81010.1093/ajcn/41.4.810)| false
McCroryM.A.GomezT.D.BernauerE.M. & MoléP.A. (1995). Evaluation of a new air displacement plethysmograph for measuring human body composition. Medicine & Science in Sports & Exercise27(12) 1686–1691. PubMed ID: 8614326 doi:10.1249/00005768-199512000-00016
McCrory, M.A., Gomez, T.D., Bernauer, E.M., & Molé, P.A. (1995). Evaluation of a new air displacement plethysmograph for measuring human body composition. 27(12), 1686–1691. PubMed ID: 8614326 doi:10.1249/00005768-199512000-0001610.1249/00005768-199512000-00016)| false
Tucker, L.A., Lecheminant, J.D., & Bailey, B.W. (2014). Test-retest reliability of the Bod Pod: The effect of mu. 118(2), 563–570. PubMed ID: 24897887 doi:10.2466/03.PMS.118k15w510.2466/03.PMS.118k15w5)| false