The purpose of this study was to conduct a comprehensive evaluation of the ActiGraph GT3X+ (AG) and activPAL (AP) for assessing time spent in sedentary behaviors (SB) in youth using structured and free-living activities. Forty-four participants (M age, 12.7±0.8 yrs) completed up to eight structured activities and approximately 2 hrs of free-living activity while wearing an AG (right hip) and AP (right thigh). A Cosmed K4b2 was used for measured energy expenditure (METy; activity VO2 ÷ resting VO2). Direct observation was used during the structured activities. SB time was estimated using the inclinometer function of the AP and AG, and count thresholds with AG (<75 vector magnitude [VM] counts/10-s; <25 vertical axis [VA] counts/10-s; and <50, 100, 150, and 200 VA counts/min). For the structured activities, the AG inclinometer and AP correctly classified supine rest about 45% of the time, seated activities 54.6% and 65.1% of the time, respectively, and walking and running >96% of the time. For the free-living measurement, the VA <25 counts/10-s had the lowest RMSE (20.6 min), while the VM <75 counts/10-s had the lowest MAPE (69.2%). The AG inclinometer was within 0.2 minutes of measured time, but had the highest MAPE (107.1%). The AP was within 1.6 minutes of measured time, but had the highest RMSE (28.5 minutes). Compared to measured SB time, the VA <25 counts/10-s and VM <75 counts/10-s provided the most precise estimates of SB during free-living activity. Further refinement is needed to improve the AP and AG posture estimates.
Scott E. Crouter, Paul R. Hibbing, and Samuel R. LaMunion
Paul R. Hibbing, Samuel R. LaMunion, Haileab Hilafu, and Scott E. Crouter
Background: Bout detection algorithms are used to segment data from wearable sensors, but it is challenging to assess segmentation correctness. The purpose of this study was to present and demonstrate the Transition Pairing Method (TPM), a new method for evaluating the performance of bout detection algorithms. Methods: The TPM compares predicted transitions to a criterion measure in terms of number and timing. A true positive is defined as a predicted transition that corresponds with one criterion transition in a mutually exclusive pair. The pairs are established using an extended Gale-Shapley algorithm, and the user specifies a maximum allowable within-pair time lag, above which pairs cannot be formed. Unpaired predictions and criteria are false positives and false negatives, respectively. The demonstration used raw acceleration data from 88 youth who wore ActiGraph GT9X monitors (right hip and non-dominant wrist) during simulated free-living. Youth Sojourn bout detection algorithms were applied (one for each attachment site), and the TPM was used to compare predicted bout transitions to the criterion measure (direct observation). Performance metrics were calculated for each participant, and hip-versus-wrist means were compared using paired t-tests (α = 0.05). Results: When the maximum allowable lag was 1-s, both algorithms had recall <20% (2.4% difference from one another, p < .01) and precision <10% (1.4% difference from one another, p < .001). That is, >80% of criterion transitions were undetected, and >90% of predicted transitions were false positives. Conclusion: The TPM improves on conventional analyses by providing specific information about bout detection in a standardized way that applies to any bout detection algorithm.
David R. Bassett, John Pucher Jr., Ralph Buehler, Dixie L. Thompson, and Scott E. Crouter
This study was designed to examine the relationship between active transportation (defined as the percentage of trips taken by walking, bicycling, and public transit) and obesity rates (BMI ≥ 30 kg · m−2) in different countries.
National surveys of travel behavior and health indicators in Europe, North America, and Australia were used in this study; the surveys were conducted in 1994 to 2006. In some cases raw data were obtained from national or federal agencies and then analyzed, and in other cases summary data were obtained from published reports.
Countries with the highest levels of active transportation generally had the lowest obesity rates. Europeans walked more than United States residents (382 versus 140 km per person per year) and bicycled more (188 versus 40 km per person per year) in 2000.
Walking and bicycling are far more common in European countries than in the United States, Australia, and Canada. Active transportation is inversely related to obesity in these countries. Although the results do not prove causality, they suggest that active transportation could be one of the factors that explain international differences in obesity rates.
Kelley Strohacker, Lindsay P. Toth, Lucas F. Sheridan, and Scott E. Crouter
Ecological momentary assessment (EMA) and accelerometer-based devices can be used concurrently to better understand dimensions of physical activity. This study presents procedures for analyzing data derived from both methods to examine exercise-related walking and running, as well as determine evidence for alignment between these methods. The participants (N = 29) wore an ActiGraph GT3X+ and completed four EMA surveys/day across 2 weeks to report exercise (mode and duration). GT3X+ counts per 10 s were processed using the Crouter two-regression model to identify periods of walking/running (coefficient of variation in activity counts ≤10% and >0%). Two reviewers visually inspected Crouter two-regression model data and recorded durations of walking/running in time blocks corresponding to EMA reports of exercise. The data were classified as “aligned” if the duration of walking/running between methods were within 20% of one another. Frequency analyses determined the proportion of aligned versus nonaligned exercise durations. Reviewer reliability was examined by calculating interobserver agreement (classification of aligned vs. nonaligned) and intraclass correlation coefficients (ICC; duration based on coefficient of variation). Of the 139 self-reported bouts of walking and running exercise, 25% were classified as aligned with the Crouter two-regression model coefficient of variation. Initial interobserver agreement was 91, and ICCs across data classified as aligned (ICC = .992) and nonaligned (ICC = .960) were excellent. These novel procedures offer a means of isolating exercise-related physical activity for further analysis. Due to the inability to align evidence in most cases, we discuss key considerations for optimizing EMA survey questions, choice in accelerometer-based device, and future directions for visual analysis procedures.
Scott E. Crouter, Diane M. DellaValle, Jere D. Haas, Edward A. Frongillo, and David R. Bassett
The purpose of this study was to compare the 2006 and 2010 Crouter algorithms for the ActiGraph accelerometer and the NHANES and Matthews cut-points, to indirect calorimetry during a 6-hr free-living measurement period.
Twenty-nine participants (mean ± SD; age, 38 ± 11.7 yrs; BMI, 25.0 ± 4.6 kg·m-2) were monitored for 6 hours while at work or during their leisure time. Physical activity (PA) data were collected using an ActiGraph GT1M and energy expenditure (METs) was measured using a Cosmed K4b2. ActiGraph prediction equations were compared with the Cosmed for METs and time spent in sedentary behaviors, light PA (LPA), moderate PA (MPA), and vigorous PA (VPA).
The 2010 Crouter algorithm overestimated time spent in LPA, MPA, and VPA by 9.0%−44.5% and underestimated sedentary time by 20.8%. The NHANES cut-points overestimated sedentary time and LPA by 8.3%−9.9% and underestimated MPA and VPA by 50.4%−56.7%. The Matthews cut-points overestimated sedentary time (9.9%) and MPA (33.4%) and underestimated LPA (25.7%) and VPA (50.1%). The 2006 Crouter algorithm was within 1.8% of measured sedentary time; however, mean errors ranged from 34.4%−163.1% for LPA, MPA, and VPA.
Of the ActiGraph prediction methods examined, none of them was clearly superior for estimating free-living PA compared with indirect calorimetry.
Natalie Jayne Taylor, Scott E. Crouter, Rebecca J. Lawton, Mark T. Conner, and Andy Prestwich
Precise measurement of physical activity (PA) is required to identify current levels and changes in PA within a population, and to gauge effectiveness of interventions.
The Online Self-reported Walking and Exercise Questionnaire (OSWEQ) was developed for monitoring PA via the Web. Forty-nine participants (mean ± SD; age = 27 ± 11.9yrs) completed the OSWEQ and International PA Questionnaire (IPAQ) short form 3 times [T1/T2/T3 (separated by 7-days)] and wore an Actigraph-GT3X-accelerometer for 7-days between T2-T3. For each measure, estimates of average MET·min·day−1 and time spent in moderate PA (MPA), vigorous PA (VPA) and moderate and vigorous PA (MVPA) were obtained.
The OSWEQ and IPAQ demonstrated test-retest reliability for MPA, VPA, and MVPA minutes and average MET·min·day−1 between T1-T2 (OSWEQ range, r = .71–.77; IPAQ range, r = .59–.79; all, P < .01). The OSWEQ and IPAQ, compared with the GT3X, had lower estimates (mean error ± 95% PI) of MVPA MET·min·day−1 by 150.4 ± 477.6 and 247.5 ± 477.5, respectively.
The OSWEQ demonstrates good test-retest reliability over 7-days and better group level estimates of MET·min·day−1 than the IPAQ, compared with the GT3X. These results suggest that the OSWEQ is a reliable and valid measure among young/working age adults and could be useful for monitoring PA trends over time.
Susan Park, Lindsay P. Toth, Scott E. Crouter, Cary M. Springer, Robert T. Marcotte, and David R. Bassett
Purpose: To examine the effect of activity monitor placement on daily step counts when monitors are worn at different positions on the wrist/forearm and the hip. Methods: Participants (N = 18) wore eight different models (four wrist and four hip models) across four days. Each day, one hip and one wrist model were selected, and four identical monitors of each model were worn on the right hip and the non-dominant wrist/forearm, respectively, during all waking hours. Step counts of each monitor were compared to the same model worn in the referent position (wrist: proximal to ulnar styloid process; hip: midline of thigh). Percent of referent steps and mean difference between observed and referent positions were computed. Significant differences in steps between positions for each method were determined using one-way repeated measures ANOVAs. For significant main effects, pairwise comparisons with Bonferroni corrections were used to determine which positions were significantly different. Results: All wrist methods showed a significant main effect for placement (p < .05) and alternate positions were 1–16% lower than the referent position. For hip methods, only the Omron HJ-325 differed across positions (p < .05), but differences were among non-referent positions and all were within ±2% of steps recorded by the referent position. Conclusions: Researchers should be aware that positions that deviate from the manufacturer’s recommended position at the wrist could influence step counts. Of all hip methods examined, the Omron had a significant placement effect which did not constitute a practical difference.
Susan Park, Lindsay P. Toth, Paul R. Hibbing, Cary M. Springer, Andrew S. Kaplan, Mckenzie D. Feyerabend, Scott E. Crouter, and David R. Bassett
It has become common to wear physical activity monitors on the wrist to estimate steps per day, but few studies have considered step differences between monitors worn on the dominant and non-dominant wrists. Purpose: The purpose of this study was to compare four step counting methods on the dominant versus non-dominant wrist using the Fitbit Charge (FC) and ActiGraph GT9X (GT9X) across all waking hours of one day. Methods: Twelve participants simultaneously wore two monitors (FC and GT9X) on each wrist during all waking hours for an entire day. GT9X data were analyzed with three step counting methods: ActiLife algorithm with default filter (AG-noLFE), ActiLife algorithm with low-frequency extension (AG-LFE), and the Moving Average Vector Magnitude (AG-MAVM) algorithm. A 2-way repeated measures ANOVA (method × wrist) was used to compare step counts. Results: There was a significant main effect for wrist placement (F(1,11) = 11.81, p = .006), with the dominant wrist estimating an average of 1,253 more steps than the non-dominant wrist. Steps differed between the dominant and non-dominant wrist for three of the step methods: AG-noLFE (1,327 steps), AG-LFE (2,247 steps), AG-MAVM (825 steps), and approached statistical significance for FC (613 steps). No significant method x wrist placement interaction was found (F(3,9) = 2.62, p = .115). Conclusion: Findings suggest that for step counting algorithms, it may be important to consider the placement of wrist-worn monitors since the dominant wrist location tended to yield greater step estimates. Alternatively, standardizing the placement of wrist-worn monitors could help to reduce the differences in daily step counts across studies.
Lindsay P. Toth, Susan Park, Whitney L. Pittman, Damla Sarisaltik, Paul R. Hibbing, Alvin L. Morton, Cary M. Springer, Scott E. Crouter, and David R. Bassett
Purpose: To examine the effect of brief, intermittent stepping bouts on step counts from 10 physical activity monitors (PAMs). Methods: Adults (N = 21; M ± SD, 26 ± 9.0 yr) wore four PAMs on the wrist (Garmin Vivofit 2, Fitbit Charge, Withings Pulse Ox, and ActiGraph wGT3X-BT [AG]), four on the hip (Yamax Digi-Walker SW-200 [YX], Fitbit Zip, Omron HJ-322U, and AG), and two on the ankle (StepWatch [SW] with default and modified settings). AG data were processed with and without the low frequency extension (AGL) and with the Moving Average Vector Magnitude algorithm. In Part 1 (five trials), walking bouts were varied (4–12 steps) and rest intervals were held constant (10 s). In Part 2 (six trials), walking bouts were held constant (4 steps) and rest intervals were varied (1–10 s). Percent of hand-counted steps and mean absolute percentage error were calculated. One sample t-test was used to compare percent of hand-counted steps to 100%. Results: In Parts 1 and 2, the SWdefault, SWmodified, YX, and AGLhip captured within 10% of hand-counted steps across nearly all conditions. In Part 1, estimates of most methods improved as the number of steps per bout increased. In Part 2, estimates of most methods decreased as the rest duration increased. Conclusion: Most methods required stepping bouts of >6–10 consecutive steps to record steps. Rest intervals of 1–2 seconds were sufficient to break up walking bouts in many methods. The requirement for several consecutive steps in some methods causes an underestimation of steps in brief, intermittent bouts.
Samantha F. Ehrlich, Amanda J. Casteel, Scott E. Crouter, Paul R. Hibbing, Monique M. Hedderson, Susan D. Brown, Maren Galarce, Dawn P. Coe, David R. Bassett, and Assiamira Ferrara
Background: This study sought to compare three sensor-based wear-time estimation methods to conventional diaries for ActiGraph wGT3X-BT accelerometers worn on the non-dominant wrist in early pregnancy. Methods: Pregnant women (n = 108) wore ActiGraph wGT3X-BT accelerometers for seven days and recorded their device on and off times in a diary (criterion). Average daily wear-time estimates from the Troiano and Choi algorithms and the wGT3X-BT accelerometer wear sensor were compared against the diary. The Hibbing 2-regression model was used to estimate time spent in activity (during periods of device wear) for each method. Wear-time and time spent in activity were compared with multiple repeated measures ANOVAs. Bland Altman plots assessed agreement between methods. Results: Compared to the diary (825.5 minutes [795.1, 856.0]), the Choi (843.0 [95% CI: 812.6, 873.5]) and Troiano (839.1 [808.7, 869.6]) algorithms slightly overestimated wear-time, whereas the sensor (774.4 [743.9, 804.9]) underestimated it, although only the sensor differed significantly from the diary (p < .0001). Upon adjustment for average daily wear-time, there were no statistically significant differences between the wear-time methods in regards to minutes per day of moderate-to-vigorous physical activity (MVPA), vigorous physical activity, and moderate physical activity. Bland Altman plots indicated the Troiano and Choi algorithms were similar to the diary and within ≤0.5% of each other for wear-time and MVPA. Conclusions: The Choi or Troiano algorithms offer a valid and efficient alternative to diaries for the estimation of daily wear-time in larger-scale studies of MVPA during pregnancy, and reduce burden for study participants and research staff.